Perseus A500 Breathing System Disassembly Assembly Guide
91 02 842 | 16.09-4 | HQ | HO | Subject to modifications | © 2016 Drägerwerk AG & Co. KGaA
Dräger Perseus® A500 Breathing System Disassembly/Assembly
Disassembly Unscrew the CLIC adapter.
Screw the CLIC adapter.
Loosen and pull out the inspiratory port.
Push in the inspiratory port and tighten it with the knurled nut. Lift off the upper part of the breathing system.
Place the upper part of the breathing system on the lower part.
Insert the cleaning disk into the adapter. Close the adapter so that it engages.
Loosen and pull out the expiration port.
Remove the cleaning disk from the adapter. Close the adapter so that it engages.
Push in the expiratory port and tighten it with the knurled nut.
Remove the flow sensors.
Turn the three levers counterclockwise about 120°.
Insert the flow sensors.
Screw off the APL valve.
Align the APL valve correctly and tighten it with the knurled nut.
Turn the three levers clockwise about 120°. Remove the valves (red and blue).
Align the valves correctly. Insert the valves.
Dräger Perseus® A500 Breathing System Reprocessing Instructions
Manual cleaning and disinfection
Machine cleaning and disinfection
Position components so that all interior spaces and surfaces are completely flushed and water can drain off freely.
Visual check *
The following parts must be free of deposits: – Incident flow grills – Non-return valves (red and blue) If required, remove deposits on the valve plates of the non-return valves with a soft cloth. If the incident flow grills are damaged, contact DrägerService®. * For machine cleaning, position each valve so that the valve cage is at the bottom.
This guideline does not replace the instruction for use.