Zeus Breathing System Disassembly Assembly Guide Aug 2016
Disassembly Assembly Guide
91 02 537 | 16.08-1 | HQ | PP | Subject to modifications | © 2016 Drägerwerk AG & Co. KGaA
Dräger Zeus® Breathing System Disassembly/Assembly
Disassembly Insert cleaning disc into the adapter. Close adapter until it snaps.
Unscrew the CLIC adapter (optional).
Screw the CLIC adapter (optional).
Loosen and pull out the inspiration port.
Check the inspiration grills. Insert and tighten the inspiration port.
Insert the flow sensors
Screw off the APL valve.
* In case the grill is damaged contact DrägerService®
Check the expiration grills.* Insert and tighten the expiration port.
Turn the screws using the key counterclockwise about 120°.
Remove the flow sensors.
Lift off the upper part of the breathing system.
Set the upper part on the lower part of the breathing system.
Remove cleaning disc from the adapter. Close adapter until it snaps.
Loosen and pull out the expiration port.
Screw in the APL valve.
Turn the screws using the key clockwise about 120°.
Dräger Zeus® Breathing System Reprocessing Instructions
Manual cleaning and disinfection
Machine cleaning and disinfection
Position components so that all interior spaces and surfaces are completely flushed and water can drain off freely.
Manual cleaning and disinfection: Please use the disinfectants listed in the Instructions for use of the Dräger flow sensors. Machine cleaning and disinfection: Recommendation: Position components so that all interior spaces and surfaces are completely flushed and water can drain off freely. Sterilization: SpiroLife® flow sensors are not suitable for plasma or radiation sterilization.
This guideline does not replace the instruction for use.